Flooring Radiant Panel Test Apparatus
EN ISO 9239-1; ASTM E648
This test method is used to measure the critical radiant flux of horizontally-mounted floor covering systems exposed to a flaming ignition source in a graded radiant heat environment, within a test chamber. It can also be used to measure this same critical radiant flux for exposed attic floor cellulose insulation.
The European Union is using this test method for fire classification of flooring products, throughout European Member States.
The radiant heat is applied by means of a gas-fuelled panel, inclined at 30°, and directed at a horizontally mounted floor covering system specimen. The radiant panel generates a radiant energy flux distribution ranging from a nominal maximum of 10.9 kW/m² to a minimum of 1.1 kW/m².
A small stainless steel pilot burner assists in specimen ignition.
The distance burned until flame-out is reached and converted, by calibration, into an equivalent critical radiant flux, in kW/m².
A smoke measuring system, according to DIN 50055, is mounted on a separate frame at the exhaust stack.
- Angled Control Rack for convenience in use, allowing observation of the apparatus and controls during equipment set-up and calibration
- Automatic ignition of the radiant panel and safety cut-out
- Hinged access to test area, via 2 position door with observation window and sliding platform
- Stainless steel hood with smoke measurement ports
- Data acquisition and analysis software.
Instrument Supplied with Software at no extra charge. Software updates provided free of charge.
Flooring Radiant Panel
Hinged access to test area, via a 2 position door with observation window and sliding platform